Product information "Double pump head for Testomat® 808 SiO2"
2x screws M3x40, M3x50
1x shaft extension
1x spacer plate
1x double pump head with hoses
1x hose extension
1x 1.5 mm Allen key
After replacement, the pump must be recalibrated in Testomat® 808 SiO2.
The adjustment is available from firmware version T808_83-003. Update the firmware of older versions before carrying out the adjustment! (serial numbers lower 253.061)
It is important to use water with a silicate content of 0 ppm for this adjustment, otherwise the adjustment of the zero point is faulty and leads to incorrect measurement results.
To ensure this, please use the silicate filter cartridge for the T808 SiO2, article 270344, specially developed for this process.
(For additional information and notes on the equipment required for the software update, see the operating instructions as well as the "Replacing the pump head in Testomat808 SiO2" document)