2 x Sight glass windows 30x33 x Flat gasket 24x2 EPDM 601 x Bottle insert for screw cap and push-fit suction tube, 100 ml bottle1 x Bottle insert for screw cap and push-fit suction tube, 500 ml bottle1 x brushes set
The cleaning brush sets are equipped with a sun head so that you can clean the measuring chamber faster and more thoroughly. In addition, the brushes absorb the dirt better thanks to their dense bristles.[#dspacer#][#dspacer#]Consist:1x #895218 cleaning brush Ø 6mm1x #895219 cleaning brush Ø 16mm
1x #895220 cleaning brush Ø 20mm
Repair and Service Case Testomat® 808Case for regular maintenance of a Testomat® 808 and on-site service with lots of content. Content: 8 3.68x1.78 O-rings8 1.78x1.78 O-rings8 4.5x1.5 O-rings8 24x2 flat gaskets1 pump head4 500ml inserts with screw cap1 100ml insert with screw cap1 cleaning brush set4 angle screw connectors6 fuses, T 0.1 A6 fuses, T 0.2 A6 fuses, T 1.0 A6 fuses, T4A6 30x3 sight glasses2 pipes, l = 53 mm2 pipes, l = 140 mm1 optical board1 SUB-D null modem cable1 USB serial adapter2 dosing needles4 hose adapters2 magnetic stirring bars8 M3x12 screws4 M3x40 screws1 LED socket1 magnetic valveDocumentation/software (1)
Contents: 1 x 37570 Solenoid valve complete Testomat® 808 2 x 37580 Insert with screw cap and suction tube 100ml bottle 1 x 37859 double pump head Testomat® 808 1 x 31972 SUB-D null modem cable, 9-pin, 2 x socket, 3m 4 x 37579 Insert with screw cap and suction tube 500ml bottle 2 x 40050 Magnetic stirring core 4 x 37538 Tube adapter Testomat® 808 4 x 40157 Angle screw-in connector, G1/8"-6 2 x 37543 Pipe, PE, D=6 d=4 L=53mm 2 x 37544 Pipe, PE, D=6 d=4 L=140mm 6 x 40170 Viewing window 30x3 6 x 31595 Fuse solder-in, TR5, T0,1A 6 x 31592 Fuse solder-in, TR5, T1,0A 6 x 31584 Fuse, solder-in, TR5, T0.2A 6 x 31585 fuse, T0,315A 8 x 33797 O-ring 3,68 x 1,78 8 x 31666 Fuse G-, 5x20mm, T4A 8 x 33777 Flat gasket 24x2 EPDM 70 4 x 33253 Screw M3x40, A2, DIN 965 8 x 33246 pan-head screw M3x12 DIN 7985H A2 8 x 11264 O-ring 4.5 x 1.5 8 x 11245 O-ring 1,78x1,78 1 x 32286 Adapter USB > Serial -length 1.1 meter 1 x 32287 Cleaning brush set 1 x 37498 USB stick with software tool for T808 and maintenance manual 2 x 37621 Dosing needle Testomat® 808 completeTranslated with www.DeepL.com/Translator (free version)
Service set contains:2 x sight glasses 30 x 33 x flat gasket 24 x 21 x insert for screw cap & suction tube 100 ml fl.1 x insert for screw cap & suction tube 500 ml fl.1 x conversion kit double pump head1 x gear motor2 x hose connectors1 x cleaning brush
Important: After replacement, the pump must be recalibrated in Testomat® 808 SiO2.
The adjustment is available from firmware version T808_83-003. Update the firmware of older versions before carrying out the adjustment! (serial numbers lower 253.061)
It is important to use water with a silicate content of 0 ppm for this adjustment, otherwise the adjustment of the zero point is faulty and leads to incorrect measurement results.
To ensure this, please use the silicate filter cartridge for the T808 SiO2, article 270344, specially developed for this process.
(For additional information and notes on the equipment required for the software update, see the operating instructions as well as the "Replacing the pump head in Testomat808 SiO2" document)
Set with all the parts, you need for the maintenance of Testomat® 808 and Testomat® 808 SIOConnection set consists of: 15 Flat seal 24 x 2 6 Sight-glass window 30 x 3 6 O-ring 3,68 x 1,78 6 O-ring 4,5 x 1,5 6 O-ring 1,78 x 1,78 1 Pipe, PE, D=6 d=4 L=53mm
Dear customers, please note that the calibration of the Testomat® 808 SiO2 can only be carried out with silicate-free water. Therefore, please use the silicate filter cartridge specially developed for this purpose when replacing the double pump head (art. no. 40395) and the associated pump adjustment.The cartridge filled with the special resin is inserted into the sampling line of the unit before the inlet and removed again after the adjustment. The pump replacement instructions describe the use of the silicate filter, storage conditions and shelf life. After the first use, the filter can be used up to 10 times within one year for balancing. After one year, the filter must be disposed of.